Jump to Section:
Questions & Answers:
Can the user walk or stand while using the Flowtron Hydroven 3 System leg garment?
Do not stand or walk while the leg garments of the Flowtron Hydroven 3 System are on the user. Do not apply or remove garments while attached to the pump or while the pump is in operation as it may damage the garment zipper. Open the zipper only after the garment is completely deflated.
Does the Flowtron Hydroven 3 System need to be used on a flat surface?
The Flowtron Hydroven 3 System pump should be placed on a flat surface during operation.
What are the cleaning instructions for the Flowtron Hydroven 3 System?
While cleaning the Flowtron Hydroven 3 System, do not spray cleaning solutions directly onto the pump. Avoid immersing electrical parts in water during the cleaning process. Never use phenolic, hypocarbonate-based disinfectants containing cationic surfactants, ammonia-based compounds, or antiseptic solutions. These will damage the surface coating. Wipe the unit with a soft cloth dampened with a mild detergent. After cleaning, wipe with an EPA-approved hospital-grade disinfectant.
How are the Flowtron Hydroven 3 System garments cleaned?
To clean Flowtron Hydroven 3 System garments, wipe down using a neutral detergent or soap powder at 104°F (40°C). Do not machine wash or immerse in water. Do not tumble dry or iron. To disinfect after cleaning, wipe the complete garment using 70% isopropyl alcohol wipe or a chlorine-releasing agent at 1000 p.p.m. available chlorine. Do not use phenol or phenol-derivative disinfectant. Rinse with clean water and air dry thoroughly. Do not exceed 120°F (51°C). Do not autoclave.
Manuals & Files:
If you have any questions please contact our customer support.